All-Time best seller YK15AB is one of our most powerful TENS & Muscle Stimulators. Specifically designed mode buttons make pain management quick and easy.
One of our most powerful TENS & Muscle Stimulator with 20 levels of adjustable intensity. Strong long-lasting relief.
Easy to read & operate buttons for each targeted body pain relief & Massage-like Modes.
Additional adjustable speed allows you to decide the most comfortable pain relief experience.
Non-stop pain relief wherever you are with just one clip.
All-Time Best Seller
Over 180,000 Units sold every year. Simple and easy to use! Safe, Non-Invasive, All- Natural and Drug free.
minutes default timer
levels of adjustable intensity
outputs with independent intensity control in A1/A2 or B1/B2
AAA battery – Uses 3 AAA Batteries
Automatically shut off when time is due
Easy to read animated backlit LCD Display
Allows 8 pads to be used at the same time
Belt clip equipped
9 custom designed modes: Neck, Shoulder, Back, Elbow, Hip, Ankle, Knee, Foot/hand, And Wrist;
6 therapeutic methods: Knead, Acupuncture, Tapping, Cupping, Scraping, and Random.
What's in the box
1 FDA cleared Class II OTC HealthmateForever YK15AB electroTherapy device (combined TENS unit and Muscle Stimulator)
8 piece self-adhesive conductive reusable gel pads with inserted pin
4 sets of electrode wires (3.5mm plug)
3 AAA batteries
1 Pads Holder
Electronic pulse massager Output Parameters:
Output pulse width: 90μS ± 15%
Pulse Frequency: <100HZ
Max output amplitude: <5v (r.m.s)
Electrode wire plug diameter 3.5 mm
Input Voltage: DC 4.5V
Rated current: ≤ 50mA
Rated Power: ≤ 0.225W
Temperature range: 0°C to 48°C
Humidity: 10% to 85%
Atmospheric Pressure: 400 hPa to 1060 hPa
Screen size: 1.75" W x 1" L
Size: 7.75" L x 2.75" W x .75" T
Weight: 4.2 oz
Patent Number: US D723, 178S
For User Manual, please email us at email@example.com.
Set up batteries.
Connect electrode wires with control unit.
1. Plug the electrode wire into the output(s) located at the top of the device. Depending on your needs, you may use one, two, three or four outputs.
2. Ensure that the electrode wires are fully inserted and secured into the output.
3. Repeat steps 1 & 2 to connect another wire.
Connect electrode pads with wires.
Place pads on your body.
Turn on the control unit and select channel.
Adjust intensity and speed.
Turn off the device.
Remove the pads from your skin.
Take off the pads from wires. Unplug wires from device.