Designed with a fully functional touchscreen, T24AB2 let you start the pain relief process in just few touches. Selected 12 targeted body pain relief modes + 12 massage-like relaxing modes. All in one.
Fully Functional Touchscreen
Customize your pain relief choices at your finger tips. Cutting-edge manufacture processes provide great user experience.
Slim & Portable
Slim and sleek body design that fits in your pocket perfectly. Relieve anywhere, anytime you want.
Massage-Like Pain Relief
Pre-programmed pain relief modes deliver massage-like sensation such as acupuncture, shiatsu, cupping, and tapping.
Safe & Fast Relief
Blocks pain receptors in the central nervous system to prevent pain signals from reaching the brain.
Multi-area Relief at the Same Time
Customize your own pain relief mode and intensity to relax different pain areas at the same time! 4 outputs allows 8 pads to be used together.
minute adjustable timer
levels of adjustable intensity
isolated outputs in channel A/B, both have 100% power
Rechargeable internal lithium battery
Adjustable mode and intensity in channel A1, A2 or channel B1, B2
Specially designed safety lock button to prevent any incautious clicks
Wireless Pad feature compatible with WI9
Tuina (Push & Pull), Acupuncture, Tapping, Guasha (Scraping), Cupping, Shiatus (Finger Press), Pressure, Back Pain, Deep Tissue Massage, Body Building, Body Building2, Combination, Shoulder, Neck, Back, Elbow, Hip, Ankle, Foot, Hand, Knee, Wrist, Reflexology, and TENS
What's in the box
1 FDA cleared Class II OTC HealthmateForever T24AB2 TENS & Muscle Stimulator
8 piece self-adhesive conductive reusable gel pads
4 sets of electrode wires (2.5mm plug)
1 USB Charger/Wall Unit (international travel charger 110V - 240V)
1 Pads Holder
1 stylus for touchscreen
2 Wireless Transmitters
4 sets of self-adhesive large oval electrode pads
Output pulse width: 90μS ± 15%
Pulse Frequency: <100HZ
Max output amplitude: <5v (r.m.s)
Input Voltage: DC5V
Rated current: ≤ 50mA
Rated Power: ≤ .25W
Screen size: 1.75" W x 3.25" L
Size: 5" L x 2.5" W x .4" T
Weight: 4.1 oz
Patent Number: US D758, 606S
For User Manual, please email us at email@example.com.
Connect electrode wires with control unit.
1. Plug the electrode wire into the device.
2. Ensure that the electrode wires are fully inserted and secured.
3. Repeat steps 1 & 2 to connect another wire.
Connect electrode pads with wires.
1. Each wire has two leads with snap-on connectors. Snap the wire connectors onto the snap on buttons on the pads.
2. Make sure at least one pair of pads is attached to the same wire.
3. Ensure that the snap on connections are fastened.
4. Remove the plastic films on the pads.
Place pads on your body.
Turn on the control unit.
Turn off the device.
Remove the pads from your skin.
Take off the pads from wires. Unplug wires from device.